Here, we present a step-by-step example for the determination of the pile-up level for the first snapshot of data obtained for GRB 091109A in PC mode. WT mode pile-up is discussed at the end of this thread.
Pile-up occurs when there is a significant probability that multiple photons registering within a given CCD frame will have overlapping charge distributions. This means that the resulting charge spatial distribution is recognised as a single event whose energy is the sum of the overlapping events (i.e., two or more soft X-ray photons can be registered as a single higher-energy photon), or a flux loss if the charge distribution has a grade outside that clasified as a true X-ray event (0-12 for PC mode; 0-2 for WT).
First, an image should be extracted for the period of interest. For a constantly bright source, all the data can be used together. If the source fades (or brightens), however, the user should identify a period of time when they believe pile-up may be present.
A quick way to estimate whether a section of data is piled-up or not is to extract a spectrum with a full circle centred on the source for a given time interval. If the count rate is ~0.5 count s-1 or above, the data should be checked for pile-up. As mentioned in the
XSELECT thread, it is usually best to use the full grade selection (0-2 for WT; 0-12 for PC) for spectral analysis. Choose the same grade selection when determining the level of pile-up as you will use for subsequent analysis.
>xselect ** XSELECT V2.2a ** > Enter session name >[xsel] xsel:ASCA >read event > Enter the Event file dir >[./] > Enter Event file list >[ ] sw00375246000xpcw3po_cl.evt Got new mission: SWIFT > Reset the mission ? >[yes] Notes: XSELECT set up for SWIFT
Time keyword is TIME in units of s
Default timing binsize = 5.0000 Setting... Image keywords = X Y with binning = 1 WMAP keywords = X Y with binning = 1 Energy keyword = PI with binning = 1 Getting Min and Max for Energy Column... Got min and max for PI: 0 1023 Got the minimum time resolution of the read data: 2.5073 MJDREF = 5.1910000742870E+04 with TIMESYS = TT Number of files read in: 1 ******************** Observation Catalogue ******************** Data Directory is: /home/work/kpa/2009/grb091109a/00375246000-xrt/ HK Directory is: /home/work/kpa/2009/grb091109a/003752460001-xrt/ OBJECT OBS_ID DATE-OBS DATAMODE 1 Burst (309. 00375246000 2009-11-09T PHOTON > filter region sw00375246000xpcw3po.reg > extract curve > filter time cursor
In this example, we are using the 20 pixel extraction region produced by the pipeline by including
cleanup=no (see the xrtpipeline thread). The first snapshot of data has been selected using the
filter time cursor command (see the main XSELECT thread).
> extract spectrum extractor v5.11 11 May 2009 Getting FITS WCS Keywords Doing file: /home/work/kpa/2009/grb091109a/00375246000-xrt/sw00375246000xpcw3po_cl.evt[regfilter("xsel_region.xsl",X,Y)] 100% completed Total Good Bad: Time Phase Grade Cut 627 354 273 0 0 0 ================================================================================ Grand Total Good Bad: Time Phase Grade Cut 627 354 273 0 0 0 in 611.78 seconds Spectrum has 354 counts for 0.5786 counts/sec ... written the PHA data Extension
This count rate is above the limit at which pile-up should be checked. Thus, the next step is to extract a full-frame image for this time and read it into
XIMAGE to determine the level of pile-up.
> clear region > extract image > save image PC1 >$ximage
As explained in the XSELECT thread, tasks such as
XIMAGE can be called from within
XSELECT by prefacing the command with a dollar sign ($). Alternatively, the user can quit
XSELECT and call
XIMAGE directly from the command line.
> read PC1.img > cpd /xtk > disp > back Calculating background: Poisson statistics assumed Background box size = 64 Background =6.3569E-04 cts/original-pixel =6.3569E-04 cts/image-pixel =6.7586E-04 cts/sqarcmin/s =1.0433E-06 cts/original-pixel/s
Here, the image has been read in and plotted. The
back command estimates the mean background over the entire image. Although this is not, strictly speaking, necessary, the next step sometimes fails if a good, full-field estimate of the background has not been determined. The other option is explicitly to state the background level if known (in terms of counts pixel-1).
The most accurate way to estimate the pile-up level is to zoom into the source and centre it in the image, to look something like this:
Then, using the command
psf/cur, the user should click in the centre of the source and then again where the PSF fades into the background. This will be between about 15 and 25 pixels (in either the X or Y direction - the programme averages in all directions) out from the centre.
>psf/cur Select center Cursor is active Pixel coordinates X= 516.15 Y= 522.04 Distance from optical axis: 27.32408333 pixels Select outer radius Cursor is active Pixel coordinates X= 513.70 Y= 506.01 radius(arcmin) = 0.629747212 BACK not specified calculating mean background from all image Using existing background calculation Background box size = 64 Background =6.3569E-04 cts/original-pixel =6.3569E-04 cts/image-pixel =6.7586E-04 cts/sqarcmin/s =1.0433E-06 cts/original-pixel/s Using average energy for PSF: 1.00000000 Total counts (original) = 346.0 Total counts (bg subtracted) = 345.5 Total counts (psf corrected) = 396.5
This will open an interactive plot window, with panels showing both the Encircled Energy Function and the PSF.
The cyan points in the PSF plot are the only ones of interest. Thus, the easiest way to avoid getting confused is to turn off the plotting of all the other datasets, as described here:
>col off 1 2 3 4 6
The XRT PSF is modelled by a King function:
PSF(r) = [ 1 + (r/rc)2 ]-β
where, for Swift, rc ~ 5.8 and β ~ 1.55 (Moretti et al., SPIE, 2005, 5898, 360).
Pile-up affects the centre of the PSF such that too few counts will be seen with respect to the King function. Out in the wings, however, the King function will fit the data. In order to determine the point at which the data and model diverge, the King function should be fitted to the outer wings and then extrapolated to the inner region to compare with the data points. Depending on how bright the source is, at least 10-15 arcsec should be excluded from the fit initially:
>rescale x 15 >model king 1 RC: VAL( 1.000 ), SIG( 0.000 ), PLO( 0.000 ), PHI( 0.000 )? 5.8, -1 2 IN: VAL( 1.000 ), SIG( 0.000 ), PLO( 0.000 ), PHI( 0.000 )? 1.55, -1 3 S0: VAL( 1.000 ), SIG( 0.000 ), PLO( 0.000 ), PHI( 0.000 )? 1 >fit Fitting group 5, from 15.0 to 39.8 Fitting 10 points in a band of 10. 5.8000002 1.5500000 1.00000000 ( -3) W-VAR= 7.617 ( -4) W-VAR= 7.617 5.8000002 1.5500000 13.635445
This will fit the King function to the outermost data (beyong 15 arcsec: rescale x 15 ignores all the data within 15 arcsec; only the data range visible in the window is used to fit the model), producing a following plot:
Now the model should be compared to the centre of the PSF, to determine the point at which it no longer fits the data. This is the radius which should be excluded when extracting spectra or light-curves in
XSELECT. It may be useful to change the y-axis scaling as well, to be able to see the plot more clearly.
rescale can be used with or without limits.
fit plot 100 smooths the model line.
>rescale x >rescale y 0.02 20 >fit plot 100
In the above plot, the model and data diverge below about 8 arcsec. Therefore, 3 or 4 pixels (1 pixel = 2.36 arcsec) should be excluded when extracting files in
XSELECT. To do this, choose an
annulus as the region shape from the menu at the top of the ds9 window, position the annulus in the field of view and double click to bring up the parameters bpx. Then type the relevant number of pixels into the inner and outer radius buttons (30 pixels is a suitable outer radius) and click generate (middle button at the bottom of the window), followed by apply (bottom left button).
Running xrtmkarf with the PSF correction set to yes will create an ARF corrected for the loss of counts caused by this annular exclusion.
In general, WT mode data should not be affected by pile-up below intensities of about 100 count s-1. See the appendix of Romano et al. (2006) for more details. Methods for compensating when pile-up does occur are discussed below.
Please note that there are many ways to investigate the level of pile-up, and the details above are only an introduction. Other methods include:
These methods are applicable to PC and WT data analysis.